Now I have the neu Problem. Look for the appropriate SCPI-commands in your manual. The same as the structure in Attachments. So try the following: This could become a lengthy explanation
|Date Added:||18 November 2005|
|File Size:||30.54 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Now I have the neu Problem. For the measurements, go back a couple of steps and select “Measurement”, there are a couple of selections to choose from. This could become a lengthy explanation How can i do with these? Here is g;ib link for the driver: You’ve got some code there only for the assuming the visa referecne is correct and you’re asking for the voltage not current.
A GPIB VISA TIME OUT ERRORS | Keysight Community
Before you finish your program you should return your instrument from the remote state in the local state. Message 8 of See attached picture for additional functions: Thank you very much for your attention. Thank for your help.
So try the following: You have to 3441a together the building blocks to get your measurement. You can find all necessary SCPI-commands there and you can just send them to the instrument by using the above mentioned VI. First I might suggest to write in german as that might be easier for you?
Start with HPA Init. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Hello Joe and 34401w, Thank for your help.
Look for the appropriate SCPI-commands in your manual. Go to the block diagram.
I never had problems with this till now. Then I can get the curve U t and I t. You can now ask the instrument for the currently measured data i. And are you sure of the syntax for the measure command?
Hope that is of any help for you. I also sometimes ignore existing drivers and create my own.
With the above named VI you first send kind of an initialization to the instrument: Message 4 of Once these are installed, open your vi. Message 1 of The same as the structure in Attachments.
34401A GPIB VISA TIME OUT ERRORS
Thank you Very much. How can I do this? Right after that you read the answer from your instrument with the GPIB Message 10 of