The new version of the driver has been tested to be backwards compatible with previous versions. Use the one appropriate for your Operating System. Is it correct that there is no function to connect the internal DMM to Abus? You can notify a user about this post by typing username. Also, after configuring in MAX, what VIs should be run if I would like to see an emulated screen of this instrument in order to control it using my visa instrument specific Labview driver. I have my MAX setup as in the attached doc. If you use the version independent ProgId in CoCreateInstance, you will not need to modify and recompile your code.
|Date Added:||8 March 2017|
|File Size:||41.6 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Here is what I did after I installed the driver: The new version of the driver has been tested to be backwards compatible with previous versions. You will also see shortcuts to the Readme file and any programming examples for this driver.
Let me know whether or not this works! If not found, 34980x installer will either abort, warn, or install the required component as appropriate. On the Keysight website is ivo ProgrammersReference with scpi commands Where is this file?
Is the Keysight A ivi driver working with IviDMM? – IVI Foundation Web Forum
I 3498a be able to point you in the right direction. First of all, not all of the matrices in a module can participate in inter-module connections.
I hope that clears things up a little bit and helps you get started on your application.
The following commands are not implemented yet SOURce: Windows Installer Version 2. So are the above two errors because I dont have the actual hardware?
Thanks for the links. See the section on installation in this document for information on what other software is required by this driver. The supported model numbers are: Message 2 of 43980a The simulation capabilities are designed primarily to support NI Switch Executive version 2. I found the following info by searching for “abus” in the IVI driver help: It would be great if I could emulate this unit though MAX and use the drivers to control this instrument.
Instrument Driver for Agilent A switch chassis and modules – MATLAB & Simulink
How to reset all positions? You may not have to do this as you may already have one configured. Ivj understand what you are saying now. When I do this I don’t get any errors: Hi Alan, this is interesting. However, when I try to run the vi and slide the action to “Connect” and click on Execute, I get the error “Path Unsupported: In the software tab of my New Driver Session I selected Aga as the software module as seen below: But the question was something else so i didnt help so far.
Actually it doesnt help as i dont want to execute a read, but just to know it.
Instrument Driver for Agilent 34980A switch chassis and modules
Additionally, if a column needs to connect to an analog bus, then it must first connect to the row attached to the desired bus. Is 34980aa the correct way of specifying the channel list ,?
NET 34908a itself is not required by this driver. Does empty channel list in configure “” mean DMM selected? But I do have a lot of experience with the Agilent and it’s various modules, including the MUX.
Unknown Chnl or RCN. People who like this. It is able to install many of the required system components, including Windows Installer 2.
You might also find it useful to ii the Help file for the phrase “Instrument-Specific Driver Overview”.