The default ASK rate value is 10 Hz. To start the application, select a UA device and click OK to establish the connection. B Pass – Temporary degradation, self recoverable. To save the waveform for future use, click select Compatible With Intuilink and click desired file name and click 9 Exiting the waveform editor. To exit the waveform editor, click. The following instructions describe the simple procedure of installing the L-mount kit and your module in the UA chassis. Page 45 Features and Functions Function Limitation If you change to a function where the maximum frequency is less than the current function, the frequency will be adjusted to the maximum value for the new function.
|Date Added:||8 January 2008|
|File Size:||22.15 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
PM is very similar to FM, but in PM, the phase of the modulated waveform aggilent varied by the instantaneous voltage of the modulating waveform. The table below shows the general environmental requirements for this instrument.
Each time the UA is plugged in, this dialog box will appear.
Index of /Companies/Agilent/UA
Comments to this Manuals Your Name. When a function is selected, the button will be illuminated. To save the waveform for future use, click select Compatible With Intuilink and click desired file name and click 9 Exiting the waveform editor.
The following figure shows an example of a PM waveform. In this mode, the UA outputs continuously when the sweep mode is selected. The following message box will appear indicating the download in progress. The duty cycle setting applies only to a Square waveform carrier.
Output Control You can disable or enable the soft front panel output control. Soft Front Panel Operation On the Mod Frequency panel as shown in the desired modulating frequency value and select the unit from the drop down list. Characteristics and Agioent In this chapter, you will observe the product characteristics and specifications.
Index of /Companies/Agilent/U2761A
At power-on, the units for output amplitude are volts peak-to-peak. Then, start drawing the desired waveform. If any damage is found, notify the nearest Agilent Sales Office immediately. Duty Cycle square Waves Features and Functions Duty Cycle Square Waves The duty cycle of a Square wave represents the amount of time per cycle that the Square wave is at a high level assuming that the waveform is not inverted.
The period of the Square waveform is equal to the specified sweep time. Or visit Agilent worldwide web at: Analog signal generator vector signal generator 66 pages.
U2761A USB Modular Function Generator
To print the manual completely, please, download it. Features and Functions Modulating Waveform Soft Front Panel Operation On the Frequency panel as shown in desired frequency value and select the unit from the drop down list. Figure Panel View Of The Frequency Section Features and Functions Function Limitations If you change to a function where the maximum frequency is less than the current function, the frequency will be adjusted to the maximum value for the new function.
For a linear sweep, the UA varies xgilent output frequency in a linear fashion agiletn the sweep.
Agilent UA USB Modular Function/ Arbitrary Waveform Generator | eBay
The following SCPI commands show a sample procedure of generating output. To enable or disable the grid reference, click. Got it, continue to print.
Table of Contents Add to my manuals Add. Don’t have an account?
The UA begins at the start ahilent, sweeps to the stop frequency, and then resets back to the start frequency. Check Your System B. Check Your System Prior to any installation or configuration, please ensure that your PC meets the following minimum system requirements.
Getting Started If you do not wish to receive similar warning messages in u2761q future, N O T E follow the instructions below.
By pressing ‘print’ button you will print only current page. To output the PM waveform, configure the settings for the carrier frequency, output amplitude, offset voltage, modulating frequency, phase deviation, and the desired waveform.